- Department of Neuroscience, Division of Neurosurgery, University of Brescia, Italy
Pier Paolo Panciani
Department of Neuroscience, Division of Neurosurgery, University of Brescia, Italy
DOI:10.4103/2152-7806.105275Copyright: © 2012 Mearini M This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
How to cite this article: Mearini M, Bergomi R, Panciani PP, Stefini R, Esposito G, Sicuri GM, Costi E, Ronchetti G, Fontanella M. Dorsal paddle leads implant for spinal cord stimulation through laminotomy with midline structures preservation. Surg Neurol Int 31-Dec-2012;3:164
How to cite this URL: Mearini M, Bergomi R, Panciani PP, Stefini R, Esposito G, Sicuri GM, Costi E, Ronchetti G, Fontanella M. Dorsal paddle leads implant for spinal cord stimulation through laminotomy with midline structures preservation. Surg Neurol Int 31-Dec-2012;3:164. Available from: http://sni.wpengine.com/surgicalint_articles/dorsal-paddle-leads-implant-for-spinal-cord-stimulation-through-laminotomy-with-midline-structures-preservation/
Background:Pain relief obtained with spinal cord stimulation (SCS) in failed back surgery syndrome (FBSS) has been shown to be more effective with paddle leads than with percutaneous catheters. A laminectomy is generally required to implant the paddles, but the surgical approach may lead to iatrogenic spinal instability in flexion. In contrast, clinical and experimental data showed that a laminotomy performed through flavectomy and minimal resection of inferior and superior lamina with preservation of the midline ligamentous structures allowed to prevent iatrogenic instability. Aim of the study was to assess degree of instability and pain level in patients operated for SCS through laminectomy or laminotomy with midline structures integrity. The surgical technique is described and our preliminary results are discussed.
Methods:Nineteen patients with FBSS underwent SCS, 12 through laminectomy and 7 through uni- or bilateral interlaminotomy with supraspinous ligament preservation. Postoperative local pain was evaluated at 15, 30, and 60 days. Static and dynamic X-rays were performed after 2 months.
Results:The techniques allowed implanting the paddle leads in all cases. No intraoperative complications occurred. Local pain was higher and recovery time was longer in patients with laminectomy. We did not observe radiological signs of postoperative iatrogenic vertebral instability. Nevertheless, two patients who underwent laminectomy showed persistence of local pain after 2 months probably due to pathologic compensatory stability provided by the paraspinal musculature.
Conclusions:The laminotomy is a minimally invasive approach that ensures rapid recovery after surgery, spinal functional integrity, and complete reversibility. Further studies are needed to confirm our preliminary results.
Keywords: Failed back surgery syndrome, laminotomy, paddle lead, spinal cord stimulation minimally invasive technique
Spinal cord stimulation (SCS) is an effective therapy in chronic intractable pain of failed back surgery syndrome (FBSS) with pain relief rates between 50% and 75% in long-term follow-up.[
The paddle lead is usually implanted through a bilateral flavectomy and partial laminectomy with midline ligamentous structures resection. In contrast, a bilateral laminotomy with midline structures preservation[
In our Neurosurgical Division SCS for FBSS was performed in 19 patients through a uni- or bilateral laminotomy or a partial laminectomy. Aim of the study was to assess if a minimally invasive approach may allow reducing spinal instability and local pain after surgery.
We presented our preliminary findings and discussed advantages and limitations of this microinvasive technique. The clinical efficacy of SCS in terms of pain relief is not debated.
Between July 2009 and December 2011, 19 patients with FBSS underwent SCS. Uni- or bilateral interlaminotomy with supraspinous ligament preservation was used in seven patients. A standard approach, through a partial laminectomy[
The technical features of paddle leads and implantable pulse generator (IPG) are summarized in
Surgical-related pain was evaluated after 15, 30, and 60 days through the numeric rating scale (NRS-11).[
The paddle leads were placed in all cases at D8-D9 level. The laminectomy was performed using a standard approach.[
The unilateral approach was preferred in case of 4 + 4-poles plates, whereas the bilateral was used for 5 + 6 + 5-poles devices.
In both cases we used the following protocol: (a) general anesthesia; (b) prone position; (c) antibiotic prophylaxis; (d) D9-D10 interlaminotomy for 4 + 4-poles electrodes and D10-D11 interlaminotomy for 5 + 6 + 5-poles plates positioning; and (e) 3 weeks of trial stimulation.
Unilateral approach [ Figure 1 ]
(a-b) Monolateral interlaminotomy with dura mater exposed (white arrow). Supraspinous ligament structures are preserved (white star). Controlateral extension of the exposure under the midline ligamentous structures (blue star), (c) Lead blank insertion through the interlaminotomy, (d) Lead paddle insertion through the interlaminotomy
We performed the unilateral approach in two cases.
(a) unilateral paraspinal muscles dissection; (b) interlaminotomy: Flavectomy and minimal resection of both inferior lamina of D9 (or D10) and superior lamina of D10 (or D11) [Figure
Bilateral approach [ Figure 3 ]
The bilateral approach was used in five cases.
(a) bilateral paraspinal muscles dissection; (b) symmetrical bilateral interlaminotomy: Bilateral flavectomy and minimal bilateral lamina resection performed as in the monolateral approach; (c) a space to introduce the lead blank and the lead paddle is obtained completely preserving the supraspinous ligament [
Surgical-related pain was divided in four levels according NRS-1:[
In all cases the paddle leads (3 Hinged 4 + 4 and 4 Specify 5 + 6 + 5) were implanted through uni- or bilateral interlaminotomy with complete preservation of the supraspinous ligament. In one case a unilateral approach was converted to a bilateral to achieve a median position of the electrode. We did not have any intraoperative problem during surgery technique related.
All patients stood up on the same day of surgery and were discharged within 48-72 hours without neurological deficits or wound problems. The postoperative X-rays showed the correct position of the plates.
After the epidural trial, we removed 1 Hinged 4 + 4 because ineffective and 1 Specify 5 + 6 + 5 because of an infection. The other five patients underwent the IPG implant (two, Synergy Versitrel – three, Prime Advanced).
After 2 months no patient complained of local pain where the interlaminotomy was performed. The radiological assessment showed that no paddle displacement occurred and no signs of segmental instability were observed [
The paddle leads were implanted with midline ligamentous structures resection. No surgical complications were reported. Ten patients were discharged within 72 hours, but 2 patients remained hospitalized 10 days for severe pain at surgical site. X-rays showed that all plates were correctly positioned. Two Hinged and 1 Specify were removed after the epidural trial because it was ineffective. Therefore, IPG implant was performed in nine patients (four, Synergy Versitrel – five, Prime Advanced).
Six patients showed moderate and severe local pain 15 days after surgery. Mild and moderate pain was still present in six cases after 30 days. Only two patients presented mild local pain after 2 months. No paddle displacement occurred and no radiological signs of instability were reported.
Literature showed that the paddle leads[
In an animal experimental model by Tai et al., a hydraulic testing machine was used to generate a 8400 N-mm increasing force in flexion and extension. The intervertebral displacement was measured after lumbar decompression obtained through bilateral laminotomy with suprapinous ligament preservation and through bilateral laminectomy with supraspinous ligament resection and was compared with an intact group. Authors showed that the lumbar spine group with a destroyed supraspinous ligament is more likely to develop instability, whereas no significant differences were found between the other two groups.[
Various studies showed that total laminectomy, both in lumbar and in thoracic spine, increases segmental instability.[
Bilateral laminotomy with resection of ligamentum flavum and superior and inferior laminar margins[
Moreover, recent experimental data[
In our experience, in case of FBSS, paddle leads for SCS were placed through standard laminectomy or interlaminotomy, which allowed to preserve the midline structures.[
Radiological signs of spinal instability were not observed. Nevertheless, two patients with laminectomy showed local pain after 2 months. The pathologic compensatory stability provided by the paraspinal musculature might be related to the persistence of pain. However, a long-term radiological and clinical follow-up is required to confirm this hypothesis.
Paddles placed through unilateral interlaminotomy tend to arrange itself in oblique cranial-lateral direction. Therefore, several attempts may be required to achieve the right position and it could be necessary to switch to a bilateral approach. Thus, we suggest the unilateral approach for small multi-column paddles (8-poles).
We consider the interlaminotomy with preservation of spinous process and supraspinous ligament for paddle leads placement a minimally invasive technique that provides a complete spinal functional integrity and reversibility. It allows achieving similar results compared with the traditional procedure with less local pain, rapid recovery after surgery, and without the risk of segmental instability.
The bilateral approach may be used in case of large paddles (16-poles), whereas the unilateral techniques should be preferred for small plate leads.
We speculate that it should be performed in all cases of paddle lead insertion for SCS, especially in the cervical spine where postlaminectomy instability is often observed.
1. Abeloos L, De Witte O, Riquet R, Tuna T, Mathieu N. Long-term outcome of patients treated with spinal cord stimulation for therapeutically refractory failed back surgery syndrome: A retrospective study. Neurochirurgie. 2011. 57: 114-9
2. Ball PA, Fanciullo GJ. Pont de dolor: A dual laminotomy technique for placing and securing an electrode in the epidural space and comments about anatomic variation that may complicate spinal cord stimulator electrode placement. Neuromodulation. 2003. 6: 92-4
3. Barolat G. Epidural spinal cord stimulation with a multiple electrode paddle leads is effective in treating intractable low back pain. Neuromodulation. 2001. 4: 59-66
4. Barolat G, Sharan A, Ong J, Simpson BA.editors. Spinal cord stimulation for back pain. Electrical stimulation and the relief of pain. Elsevier, Pain Res and Clin Man; 2003. 15: 79-86
5. Cameron T. Safety and efficacy of spinal cord stimulation for the treatment of chronic pain: A 20-year literature review. J Neurosurg. 2004. 100: 254-67
6. Chen LH, Lai PL, Tai CL, Niu CC, Fu TS, Chen WJ. The effect of interspinous ligament integrity on adjiacent segment instability after lumbar instrumentation and laminectomy: An experimental study in porcine model. Biomed Mater Eng. 2006. 16: 261-7
7. Chen WJ, Lai PL, Tai CL, Chen LH, Niu CC. The effect of sagittal alignment on adjacent joint mobility after lumbar instrumentation: A biomechanical study of lumbar vertebrae in a porcine model. Clin Biomech (Bristol, Avon). 2004. 19: 763-8
8. Deutsch H, Haid RW, Rodts GE, Mummaneni PV. Postlaminectomy cervical deformity. Neurosurg Focus. 2003. 15: E5-
9. Frey ME, Manchikanti L, Benyamin RM, Shultz DM, Smith HS, Cohen SP. Spinal cord stimulation for patients with failed back surgery syndrome: A systematic review. Pain Physician. 2009. 12: 379-97
10. Hartrick CT, Kovan JP, Shapiro S. The numeric rating scale for clinical pain measurement: A ratio measure?. Pain Pract. 2003. 3: 310-6
11. Kanayama M, Abumi K, Kaneda K, Tadano S, Ukai T. Phase lag of the intersegmental motion in flexion-extension of the lumbar spine: An in vivo study. Spine (Phila Pa 1976). 1996. 21: 1416-22
12. Kumar K, Taylor RS, Jacques L, Eldabe S, Meglio M, Molet J. The effects of spinal cord stimulation in neuropathic pain are sustained: A 24-months follow-up the prospective randomized controlled multicenter trial of the effectiveness of spinal cord stimulation. Neurosurgery. 2008. 63: 762-70
13. Leveque JC, Villavicencio AT, Bulsara KR, Rubin L, Gorecki JP. Spinal cord stimulation for failed back surgery syndrome. Neuromodulation. 2001. 4: 1-9
14. Lu WW, Luk KD, Ruan DK, Fei ZQ, Leong JC. Stability of the whole lumbar spine after multilevel fenestration and discectomy. Spine (Phila Pa 1976). 1999. 24: 1277-82
15. Mayer TG, Vanharanta H, Gatchel RJ, Mooney V, Barbes D, Judge L. Comparison of CT scan muscle measurements and isokinetic trunk strength in postoperative patients. Spine (Phila Pa 1976). 1989. 14: 33-6
16. North RB, Kidd D, Shipley J, Taylor RS. Spinal cord stimulation versus reoperation for failed back surgery syndrome: A cost effectiveness and cost utility analysis based on a randomized, controlled trial. Neurosurgery. 2007. 61: 361-9
17. North RB, Kidd DH, Olin JC, Sieracki JM. Spinal cord stimulation electrode design: A prospective, randomized, controlled trial comparing percutaneous and laminectomy electrodes – part I: Technical outcomes. Neurosurgery. 2005. 51: 381-90
18. North RB, Kidd DH, Petrucci L, Dorsi MJ. Spinal cord stimulation electrode design: A prospective, randomized, controlled trial comparing percutaneous and laminectomy electrodes part II: Clinical outcomes. Neurosurgery. 2005. 57: 990-6
19. Oda I, Abumi K, Lü D, Shono Y, Kaneda K. Biomechanical role of the posterior elements, costovertebral joints, and rib cage in the stability of the thoracic spine. Spine (Phila Pa 1976). 1996. 21: 1423-9
20. Postacchini F, Cinotti G, Perugia D, Gumina S. The surgical treatment of central lumbar stenosis. Multiple laminotomy compared with total laminectomy. J Bone Joint Surg Br. 1993. 75: 386-92
21. Rigoard P, Delmotte A, Dhoutaud S, Misbert L, Diallo B, Roy-Moreau A. Back pain: A real target for spinal cord stimulation?. Neurosurgery. 2012. 70: 574-85
22. Sears NC, Machado AG, Nagel SJ, Deogaonkar M, Stanton-Hicks M, Rezai AR. Long-term outcomes of spinal cord stimulation with paddle leads in the treatment of complex regional pain syndrome and failed back surgery syndrome. Neuromodulation. 2011. 14: 312-8
23. Sharma M, Langrana NA, Rodriguez J. Role of ligaments and facets in spinal stability. Spine (Phila Pa 1976). 1995. 20: 887-900
24. Slavin KV. Letters to the editor. Neuromodulation. 2003. 6: 270-
25. Slavin KV, Burchiel KJ.editors. Two-level partial laminotomy for implantation of paddle spinal cord stimulation electrodes. Book of Abstracts, 12th World Congress of Neurosurgery. Sydney. Australia: he World Federation of Neurosurgical Societies (WFNS); 2001. p. 164-
26. Smith TH. The use of a quadruped as an in vivo model for the study of the spine: Biomechanical consideration. Eur Spine J. 2002. 11: 137-44
27. Tai CL, Hsieh PH, Chen WP, Chen LH, Chen WJ, Lai PL. Biomechanical comparison of lumbar spine instability between laminectomy and bilateral laminotomy: An experimental study in porcine model. BMC Muskoloskelet Disord. 2008. 9: 84-93
28. Taylor RS, Van Buyten JP, Buchser E. Spinal cord stimulation for chronic back and leg pain and failed back surgery syndrome: A systematic review and analysis of prognostic factors. Spine (Phila Pa 1976). 2005. 30: 152-60
29. Villavicencio AT, Leveque JC, Rubin L, Bulsara K, Gorecki JP. Laminectomy versus percutaneous electrode placement for spinal cord stimulation. Neurosurgery. 2000. 46: 399-405
30. Weiner BK, Walker M, Brower RS, McCoulloch JA. Microdecompression for lumbar spinal canal stenosis. Spine (Phila Pa 1976). 1999. 24: 2268-72
31. Yoganandan N, Maiman DJ, Pintar FA, Bennett GJ, Larson SJ. Biomechanical effects of laminectomy on thoracic spine stability. Neurosurgery. 1993. 32: 604-10